Korean J Lab Med. Jun;27(3) [Evaluation of the Abbott Cell-Dyn Sapphire hematology analyzer]. [Article in Korean]. Park Y(1), Song J, Song S. Name of instrument, CELL-DYN Sapphire*. First year installed in U.S./Outside U.S./No. of units sold in , //—. No. units installed in U.S./Outside. The CELL-DYN Sapphire hematology system is designed to address routine testing needs of high-volume hematology laboratories and difficult.
|Published (Last):||19 October 2012|
|PDF File Size:||10.31 Mb|
|ePub File Size:||12.46 Mb|
|Price:||Free* [*Free Regsitration Required]|
Abbott CELL-DYN Sapphire Hematology System
APolymorphonuclear cells. Yet, several attempts to use automated hematology analyzers for this purpose have been described over the years. Inquire Here Power Supply: The history of Abbott’s Diagnostics Division is an impressive record of innovation. Minimal maintenance with on board automated logs to track every maintenance action.
November 23 rd0 Replies Post a Reply.
Literature Review and Comparison of Clinicopathologic Features. The positive and negative likelihood were respectively 3. Inquire Here Sample Size Max: Different cell concentrations were prepared, and 10 aliquots of each cell concentration were counted.
It is a multi-parameter automated hematology analyzer designed for in vitro diagnostic use in counting and characterizing blood cells.
Secure We only pass your details on to trusted suppliers at your request. The analyzer features MAPSS technology plus 3 color fluorescence to provide our most comprehensive testing capability. Inquire Here Point Of Care: The solid lines represent the power regression lines, and the equations with corresponding correlation coefficients are the following: Cell Dyn Number of Assays: More about Abbott Diagnostics. Likewise, a very good negative likelihood ratio 0.
The Cell-Dyn Sapphire showed good sensitivity Inquire Here Reagent Type: The parameters of the regression equation are summarized in Table 2. Fast Your inquiry will be delivered straight to the manufacturer.
Cell-Dyn from Cell Dyn Sapphire. The names and logos of manufacturers, their instruments, and their products referred to herein may be protected by trademark or other law, and are used herein solely for purpose of reference. Inquire Here Reagent Delivery: Why request a quote through SelectScience? We investigated whether these limits can be modified to make them applicable for body fluids with much lower cell concentrations compared with blood.
All samples were measured twice, and the means were compared with the expected cell counts. Passing-Bablok regression analysis indeed showed lack of agreement between both methods in these samples. The parameters of the regression equation are summarized in Table 3.
Inquire Here Reagent System: For method comparison, we used CSF, serous fluid 61 pleural and 40 peritonealand CAPD fluid samples that were collected for routine diagnostic purposes.
The Cell-Dyn Sapphire flagged 7 samples, but in only 2 of those samples were metastatic cells present according to the differentiation of the cytocentrifuged preparation. Oxford University Press is a department of the University abbotr Oxford. First, all requested tests were performed. Please could you kindly explain in brief the differences between the two Machines For assessing the differential count, the percentages of lymphocytes and monocytes determined by the Cell-Dyn Sapphire were added, and this sum was compared with the mononuclear cell MNC count found on the cytocentrifuged preparations, which included lymphocytes, monocytes, and, in some samples, epithelial cells and macrophages.
Similarly, an excellent positive likelihood ratio Inquire Here Sample Size Min: Experienced observers examined at least nucleated sappire. We appreciate the support of individuals and companies making their documents and resources available to our community. The Cell-Dyn Sapphire is an automated hematology analyzer designed sappire counting peripheral blood cells in whole blood samples.
[Evaluation of the Abbott Cell-Dyn Sapphire hematology analyzer].
Receive your quote directly from the manufacturer. Automated counting of white and red blood cells in the cerebrospinal fluid. Because hematology analyzers are normally running blood samples with inherently much higher cell concentrations, the accepted background cell concentration and the elimination of carryover are important issues. Performance evaluation of the application of body fluids on the Sysmex XE series automated haematology analyzer.
The enumeration and identification of blood cells in body fluids offers important information for the diagnosis and treatment of various medical conditions. Abstract The enumeration and identification of blood cells in body fluids offers important information for the diagnosis and treatment of various medical conditions. CPolymorphonuclear cells with the cytocentrifuged preparation possibly including epithelial cells and macrophages.
DMononuclear cells. It furthers the University’s objective of excellence in research, scholarship, and education by publishing worldwide. Linearity of the automated method was assessed by using linear regression analysis.
Solid lines are the regression lines and dotted lines, the identity lines. October 6 th1 Reply Post a Reply. Agreement between the automated cell counts and the microscopic data was examined by Passing-Bablok regression analysis. Statistical analyses were performed with Analyse-it software, version 1.
Thank you very much.